Comparison of manual and mechanical cardiopulmonary resuscitation on the move using a manikin: a service evaluation
Methods: Ten practising paramedics each volunteered to participate in four pre-hospital CPR scenarios. The MCPR device used for this study was the LUCASTM2. Data were captured electronically using QCPRTM wireless technology (Resusci Anne® Wireless SkillReporterTM manikin and software by Laerdal Medical©).
Results: A reduction in the rate, depth and percentage of correct compressions was noted when the paramedics were moving the patient. In relation to the 2010 ERC guidelines, the SCPR did not meet current guidelines and was of more variable quality than MCPR. MCPR was consistent and conformed to the guidelines. However, the application of the LUCASTM2 when only one paramedic was present resulted in a significant delay in commencing chest compressions.
Conclusion: In the pre-hospital setting, transporting a patient during a cardiac arrest can have a deleterious effect on the quality of chest compressions being provided. When provided by a mechanical device rather than manually, the quality of chest compressions produced is closer to that currently recommended, but two persons would be required for timely deployment of the device and to maximise the chest compression fraction. This could suggest a potential use for pre-hospital MCPR even in the absence of recommendation for routine use.
Document Type: Research Article
Affiliations: 1: Email: [email protected] 2: Email: [email protected] 3: Email: [email protected] 4: Email: [email protected] 5: Email: [email protected] 6: Email: [email protected] 7: Email: [email protected] 8: Email: [email protected]
Publication date: December 1, 2017
- The BPJ is a quarterly electronic journal committed to publishing high-quality research and increasing the evidence-base for the paramedic profession. It is freely available to all members of The College of Paramedics.
- Editorial Board
- Information for Authors
- Submit a Paper
- Membership Information
- Terms & Conditions
- Ingenta Connect is not responsible for the content or availability of external websites
- Access Key
- Free content
- Partial Free content
- New content
- Open access content
- Partial Open access content
- Subscribed content
- Partial Subscribed content
- Free trial content