PRECISION OF OLFACTOMETRY AND ODOR TESTING RESULTS
Abstract:Olfactometry is a precise testing method. How precise depends on the quality assurance / quality control (QA/QC) statistics associated with the laboratory's results. The European Olfactometry testing standard, EN13725:2003, provides a formalized method for monitoring the performance of panel members (assessors) and test results. This is accomplished through testing with the standard odorant 1- butanol (n-butanol).
Olfactometry precision can be understood through four different variance values: 1. specific panelvariance, 2. within panel variance, 3. inter-panel variance, and 4. inter-laboratory variance.
Inter-laboratory studies have shown standard deviation of log threshold values as high as 0.30 andindividual laboratories have shown inter-panel standard deviations of 0.10 and within panel standard deviations as low as 0.05. This paper defines the four types of variances and shows how the precision of odor panel results can be expressed as analogous to a group of assessors determining sound level within ±1dB in 95% of the cases.
This paper will present these different measures of precision with test data and examples of applications of how this information can be utilized by various stakeholders.
Document Type: Research Article
Publication date: January 1, 2006
More about this publication?
- Proceedings of the Water Environment Federation is an archive of papers published in the proceedings of the annual Water Environment Federation® Technical Exhibition and Conference (WEFTEC® ) and specialty conferences held since the year 2000. These proceedings are not peer reviewed. WEF Members: Sign in (right panel) with your IngentaConnect user name and password to receive complimentary access.
- Subscribe to this Title
- Membership Information
- About WEF Proceedings
- WEFTEC Conference Information
- Ingenta Connect is not responsible for the content or availability of external websites