Image Curvature Correction and Cosmic Removal for High-Throughput Dispersive Raman Spectroscopy
Abstract:A key factor determining the sensitivity of a Raman spectrometer is the usable detection area, which is the product of the usable slit width and the height. For the majority of process Raman samples, the larger the sampling area is, the more the scattered Raman signal can be gathered. On a multi-channel-detector-based dispersive spectrometer, a given spectral resolution limits the slit width. Extending the slit height using a straight slit usually causes the image to be curved on the detector due to optical effects. If left untreated, the curved slit image will degrade the peak shape and spectral resolution; therefore, the slit height must also be kept small if this negative effect is to be avoided. The mechanism of the curvature formation was analyzed for an on-axis-lens-based spectrograph, and a correction technique was developed to generate a straight slit image on the charge-coupled device (CCD). This allowed a large portion of the CCD height to be used without degrading the spectral resolution. A large fiber bundle was usable instead of a single small core fiber, generating significant increase in collected signal strength in clear or translucent samples. The straight image also enabled a new cosmic spike removal method, wherein the CCD image was divided into multiple strips, and a comparison among them allowed the identification and removal of cosmic spikes in a single CCD integration. On the contrary, many existing cosmic removal methods rely on comparison of multiple sequentially acquired spectra, potentially introducing artifacts, particularly when the spectral features are changing.
Document Type: Research Article
Publication date: November 1, 2003
More about this publication?
- The Society publishes the internationally recognized, peer reviewed journal, Applied Spectroscopy, which is available both in print and online. Subscriptions are included with membership or can be purchased by institutional or corporate organizations. Abstracts may be viewed free of charge. Previously published as Bulletin (Society for Applied Spectroscopy)
- Editorial Board
- Information for Authors
- Submit a Paper
- Subscribe to this Title
- Membership Information
- Request copyrighted SAS materials
- Spectroscopic Nomenclature
- Focal Point (Open Access)
- ingentaconnect is not responsible for the content or availability of external websites