Remote Elemental Analysis by Laser-Induced Breakdown Spectroscopy Using a Fiber-Optic Cable
Abstract:The elemental composition of solids can be determined rapidly and simply with the use of laser-induced breakdown spectroscopy (LIBS). This method, described in detail elsewhere, uses powerful laser pulses to form a microplasma or spark on a sample. A small amount of material is vaporized, and emitting species in the plasma are identified by spectrally and temporally resolving the spark light. Although LIBS measurements can be performed remotely on solids at distances up to 24 m from the laser and detection system with a long-focal-length lens, this method has some disadvantages including safety (the possibility of ocular damage by the high-energy laser pulses), need for a clear line of sight to the analysis area, scattering of incident pulse energy by dusts or fogs, and problems associated with precise focusing of laser beams at long distances. In particular, the plasma will preferentially form on dust particles in front of the sample because of the long Rayleigh length of the focused beam.
Document Type: Short Communication
Affiliations: Chemical Science and Technology Division, Los Alamos National Laboratory, Los Alamos, New Mexico 87545
Publication date: June 1, 1995
More about this publication?
- The Society publishes the internationally recognized, peer reviewed journal, Applied Spectroscopy, which is available both in print and online. Subscriptions are included with membership or can be purchased by institutional or corporate organizations. Abstracts may be viewed free of charge. Previously published as Bulletin (Society for Applied Spectroscopy)
- Editorial Board
- Information for Authors
- Submit a Paper
- Subscribe to this Title
- Membership Information
- Request copyrighted SAS materials
- Spectroscopic Nomenclature
- Focal Point (Open Access)
- ingentaconnect is not responsible for the content or availability of external websites