We have investigated a new pulsed light source for time-resolved fluorimetry. The light source consists of a beta emitter, Sr90, and a liquid or plastic scintillator. Time-correlated single photon counting is employed to demonstrate the new pulsed light source. Measured fluorescence
lifetimes of standards anthracene (4.97 ± 0.06 ns) and 1-aminoanthracene (21.4 ± 0.2 ns) agree well with literature values. Advantages of the new light source include the low cost, the size (1 cm-3 or smaller), the nanosecond pulse widths, the high pulse rates, and
the extremely stable pulse-to-pulse profile. There is no risk of radioactive contamination from the beta emitter since it is bound in a ceramic matrix and encapsulated.
Department of Chemistry, Indiana University, Bloomington, Indiana 47405
Publication date: January 1, 1995
More about this publication?
The Society publishes the internationally recognized, peer reviewed journal, Applied Spectroscopy, which is available both in print and online. Subscriptions are included with membership or can be purchased by institutional or corporate organizations. Abstracts may be viewed free of charge. Previously published as Bulletin (Society for Applied Spectroscopy)