Experimental Alignment Using Reverse-Path Laser Imaging
Abstract:In many research laboratories, modular spectrometric components are used. If experiments are often changed, alignment can be time consuming and difficult. In our atomic emission spectrometry (AES) laboratory, the light source and the spectrometer are not physically connected together, and the spectrometer is mobile—which necessitates frequent checking of the focusing and the alignment of the system. Previously we used a lens to collect light from our inductively coupled plasma for AES experiments. The focusing and alignment had been done by observing the plasma image on the entrance slit. While this approach is visually appealing, chromatic aberrations will inevitably cause defocusing in the ultraviolet region of the spectrum, and it is critical that we be able to image properly in this region. Recently we switched to the use of an off-axis concave mirror to eliminate chromatic aberrations caused by the use of the lens. It became evident during the design phase of this conversion that alignment would become more challenging due to the off-axis arrangement required by this configuration (Fig. 1).
Document Type: Research Article
Affiliations: Department of Chemistry, McGill University, 801 Sherbrooke St. W., Montreal, Quebec H3A 2K6, Canada
Publication date: November 1, 1993
More about this publication?
- The Society publishes the internationally recognized, peer reviewed journal, Applied Spectroscopy, which is available both in print and online. Subscriptions are included with membership or can be purchased by institutional or corporate organizations. Abstracts may be viewed free of charge. Previously published as Bulletin (Society for Applied Spectroscopy)
- Editorial Board
- Information for Authors
- Submit a Paper
- Subscribe to this Title
- Membership Information
- Request copyrighted SAS materials
- Spectroscopic Nomenclature
- Focal Point (Open Access)
- ingentaconnect is not responsible for the content or availability of external websites