If you are experiencing problems downloading PDF or HTML fulltext, our helpdesk recommend clearing your browser cache and trying again. If you need help in clearing your cache, please click here . Still need help? Email email@example.com
In many high-pressure vibrational spectroscopic studies the pressure required is below about 10 kbar, Moreover, infrared microanalysis is usually carried out with a diamond anvil cell under a pressure of less than 1 kbar. Therefore, we have recently developed a cheaper and more readily
available glass anvil cell for high-pressure Raman and infrared spectroscopies. However, glass strongly absorbs infrared radiation in the frequency region below 2750 cm−1 and thus the glass anvil cell can be used for high-pressure infrared spectroscopy only in the frequency
range above 2750 cm−1.
Division of Chemistry, National Research Council of Canada, Ottawa, Ontario K1A 0R6, Canada
Publication date: July 1, 1984
More about this publication?
The Society publishes the internationally recognized, peer reviewed journal, Applied Spectroscopy, which is available both in print and online. Subscriptions are included with membership or can be purchased by institutional or corporate organizations. Abstracts may be viewed free of charge. Previously published as Bulletin (Society for Applied Spectroscopy)