A Step-Wise Variable Path Far-Infrared Solution Cell for Use in Vacuum
Abstract:Far-infrared spectroscopy (particularly at resolutions of 2 cm−1 or better) is complicated by the presence of sharp, strong H2O vapor absorption lines. One common technique for reducing the effect of these lines is to use windows to isolate the sample chamber from the rest of the spectrometer. The sample chamber is then flushed vigorously with dry air or nitrogen, while the rest of the spectrometer is either evacuated or continuously flushed, depending on its construction. The lines due to residual H2O are expected to cancel when the reference and sample spectra are compared. With our instrument (a Digilab FTS 16 interferometer, which can be evacuated) we have achieved better cancellation more quickly by removing the windows from the sample compartment. The entire instrument is then evacuated for each sample. The time required for evacuation is quite short (about 5 min) and when many different sample spectra are to be compared to the same reference spectrum (stored in the computer) it is advantageous not to have to worry about changes in flushing rate or time or in the degree of dryness of the air used for flushing. Also, by removing the windows between the sample compartment and the rest of the interferometer, the signal/noise ratio is improved.
Document Type: Research Article
Affiliations: Department of Chemistry, University of Calgary, Calgary, Alberta, Canada T2N 1N4
Publication date: March 1, 1978
More about this publication?
- The Society publishes the internationally recognized, peer reviewed journal, Applied Spectroscopy, which is available both in print and online. Subscriptions are included with membership or can be purchased by institutional or corporate organizations. Abstracts may be viewed free of charge. Previously published as Bulletin (Society for Applied Spectroscopy)
- Editorial Board
- Information for Authors
- Submit a Paper
- Subscribe to this Title
- Membership Information
- Request copyrighted SAS materials
- Spectroscopic Nomenclature
- Focal Point (Open Access)
- ingentaconnect is not responsible for the content or availability of external websites