A computerized direct reading emission spectrometer has been constructed to perform rapid spectrochemical analyses. The four essential components include a prism-echelle spectrograph to disperse light from the source, a proximity-focus image intensifier to convert ultraviolet radiation
to visible, a random-access digital television camera to measure the intensities of many wavelengths simultaneously, and a dedicated minicomputer for control, data processing and output. At the present stage of development, the intensities of 400 analytical lines can be measured every second
during the excitation of a sample and the results can be summed, averaged, or displayed as intensity vs time. Analytical working curves can be automatically constructed and displayed for inspection by the operator. Analyses are reported for Fe in beryl using a graphite powder method and a
dc arc, as well as for Zn, Sn, and Pb in Cu alloys using a glow-discharge sputtering source. The analyses yield values comparable to those from more conventional equipment.
Bell Laboratories, Murray Hill, New Jersey 07974
Publication date: July 1, 1975
More about this publication?
The Society publishes the internationally recognized, peer reviewed journal, Applied Spectroscopy, which is available both in print and online. Subscriptions are included with membership or can be purchased by institutional or corporate organizations. Abstracts may be viewed free of charge. Previously published as Bulletin (Society for Applied Spectroscopy)