Simple Helium Enclosure for Increasing Diffraction-Line Detectability with a G.E. XRD-5 Diffractometer
Abstract:Various means may be used to improve the detectability of an x-ray diffractometer, such as using pulse-height selection, etc. One method of increasing detectability is that of using a helium path instead of an air path for the x-rays. A graph comparing the detectability of diffraction-line intensities in a helium path to that in an air path for various wavelengths is given in Fig. 1. Chromium K alpha (CrKα) radiation is used for greater dispersion of large d spacings such as are encountered in many organic compounds. The graph in Fig. 1. indicates that the detectability for CrKα is increased approximately 2½ times by using a helium path instead of an air path. A description follows for a simple method of constructing a helium enclosure for the G.E. XRD-5 diffractometer.
Document Type: Short Communication
Affiliations: X-Ray Crystallography Laboratory, UNIVAC, St. Paul, Minnesota
Publication date: May 1, 1968
More about this publication?
- The Society publishes the internationally recognized, peer reviewed journal, Applied Spectroscopy, which is available both in print and online. Subscriptions are included with membership or can be purchased by institutional or corporate organizations. Abstracts may be viewed free of charge. Previously published as Bulletin (Society for Applied Spectroscopy)
- Editorial Board
- Information for Authors
- Submit a Paper
- Subscribe to this Title
- Membership Information
- Request copyrighted SAS materials
- Spectroscopic Nomenclature
- Focal Point (Open Access)
- ingentaconnect is not responsible for the content or availability of external websites