Automatic Controller and Data Printer For Use With a Philips Three Position Spectrograph
Abstract:This report describes equipment that converts a Philips. Three Position X-Ray Spectrograph to semiautomatic operation for routine sample analysis. Up to ten independent sets of conditions for wave-length (2Θ), pulse height analysis baseline, and total count can be selected in advance. The equipment will then automatically determine the "counting time" for a sample for each set of conditions selected. At the end of each count the equipment prints out the sample number, code numbers for the goniometer position, baseline setting, the fixed count used, and the time, in tenths of a sec, required to reach the fixed count. Semiautomatic operation permits operation by personnel with minimum training, reduces invalid results due to operator mistakes, and increases sample output per day. The results are identical with those obtained by manual operation.
Document Type: Research Article
Affiliations: Division of Color & Cosmetics, Food and Drug Administration. U. S. Department of Health, Education, and Welfare, Washington, D. C.
Publication date: July 1, 1963
More about this publication?
- The Society publishes the internationally recognized, peer reviewed journal, Applied Spectroscopy, which is available both in print and online. Subscriptions are included with membership or can be purchased by institutional or corporate organizations. Abstracts may be viewed free of charge. Previously published as Bulletin (Society for Applied Spectroscopy)
- Editorial Board
- Information for Authors
- Submit a Paper
- Subscribe to this Title
- Membership Information
- Request copyrighted SAS materials
- Spectroscopic Nomenclature
- Focal Point (Open Access)
- ingentaconnect is not responsible for the content or availability of external websites