Careful examination of organic liquids and solutions at very narrow spectral slit widths indicates that many compounds exhibit several absorptions of half band width narrower than 5 cm−1, particularly at the medium and long wavelengths. This suggests that sodium chloride
prism spectrophotometers are currently being operated under conditions which produce rather significant errors in intensity and band width on some narrow bands. In order that this "finite slit error ' be 3% or less in measuring peak absorbance, a spectral slit width about ⅕ of the
natural half band width must be employed, e.g., a resolution of 1 cm−1 for a band of 5 cm−1 half band width. The high resolution afforded by the grating instrument allows accurate intensity measurements on the typically narrow bands of such materials as indene
at 1018 cm−1, cyclohexane at 861 and 903 cm−1 and acetonitrile at 2255 cm−1. Other advantages of the high resolution grating spectrophotometer for solution spectroscopy are discussed.
Beckman Instruments, Inc., Fullerton, California
Publication date: January 1, 1962
More about this publication?
The Society publishes the internationally recognized, peer reviewed journal, Applied Spectroscopy, which is available both in print and online. Subscriptions are included with membership or can be purchased by institutional or corporate organizations. Abstracts may be viewed free of charge. Previously published as Bulletin (Society for Applied Spectroscopy)