Continuous Reference Beam Attenuator for Infrared Spectrophotometry
Abstract:In the course of infrared studies on polymers and other materials, one is often interested in weak absorption bands that may be significant in the structural determination of the sample. In order to increase the intensity of these bands, the operator generally increases the sample thickness and thereby frequently decreases the background transmission. However, some samples such as certain mulls or samples containing inorganic fillers or even inorganic samples themselves are quite dense and require the attenuation of the reference beam in order to record the spectrum properly. Thus it is often desirable to compensate for the fixed energy losses in the sample beam (or in effect, to expand the scale by attenuation of the reference beam). This has been accomplished through the use of wire mesh screens or slotted shutters. These methods are inconvenient since they either produce a "step-by-step" attenuation, have only a limited attenuation range, or the degree of attenuation can not be easily adjusted. A reference beam attenuator that will compensate the energy losses in the sample beam from 0 to 100% transmission in a continuous manner and can be adjusted to energy differences as small as 1% is shown in Figure 1. This attenuator was constructed for use with the Perkin-Elmer Model 21 Spectrophotometer and its mounting is shown in Figure 2.
Document Type: Research Article
Affiliations: Bell Telephone Laboratories, Inc., Muray Hill, New Jersey
Publication date: February 1, 1960
More about this publication?
- The Society publishes the internationally recognized, peer reviewed journal, Applied Spectroscopy, which is available both in print and online. Subscriptions are included with membership or can be purchased by institutional or corporate organizations. Abstracts may be viewed free of charge. Previously published as Bulletin (Society for Applied Spectroscopy)
- Editorial Board
- Information for Authors
- Submit a Paper
- Subscribe to this Title
- Membership Information
- Request copyrighted SAS materials
- Spectroscopic Nomenclature
- Focal Point (Open Access)
- ingentaconnect is not responsible for the content or availability of external websites