Arc-Spark Source Unit Changes
Abstract:In commercial d.c. arc power supplies using a motor-driven variable reactor; it is not possible to predict accurately the excitation current before the arc is started. To permit pre-arc setting of the current, a circuit (Figure 1) is in use which temporarily shorts the d.c. so that its level can be adjusted. The circuit requires two relays which are energized by a single switch. One relay pulls in fast and drops out slowly; the other pulls in slowly and drops out fast. The fast-in, slow-out relay R1 is used to disconnect the normal output and timing circuits and to short the d.c. output internally. The slow-in, fast-out relay R2 connects the primary input power to the rectifier circuit. The relays can be time-delay types using hydraulic or pneumatic principles for delay, or they can be fed through RC networks as shown in Figure 1.
Document Type: Short Communication
Affiliations: Goodyear Atomic Corporation, Portsmouth, Ohio, Pre-Arc Current Setter
Publication date: August 1, 1958
More about this publication?
- The Society publishes the internationally recognized, peer reviewed journal, Applied Spectroscopy, which is available both in print and online. Subscriptions are included with membership or can be purchased by institutional or corporate organizations. Abstracts may be viewed free of charge. Previously published as Bulletin (Society for Applied Spectroscopy)
- Editorial Board
- Information for Authors
- Submit a Paper
- Subscribe to this Title
- Membership Information
- Request copyrighted SAS materials
- Spectroscopic Nomenclature
- Focal Point (Open Access)
- ingentaconnect is not responsible for the content or availability of external websites